A Novel Detection System Consisting of a Large-Area Sensor and a Multicell Si-Pad Array Operated in Spectroscopic Mode for X-Ray Breast Imaging
Darambara, D G, Sellin, P J and Maehlum, G (2004) A Novel Detection System Consisting of a Large-Area Sensor and a Multicell Si-Pad Array Operated in Spectroscopic Mode for X-Ray Breast Imaging In: IEEE Transactions on Nuclear Science.
The ability of coherent X-ray scatter to provide the molecular structure of breast tissues could add a new dimension in X-ray breast imaging capable of tracking the molecular structural changes during disease progression and of improving the sensitivity to low-contrast lesions without increasing the radiation dose. Work is under way to build a laboratory prototype dual-sensor breast-imaging scanning system, which combines the diagnostic information from both the transmitted primary and the forward scattered X-rays. This required the design and development of a coherent X-ray scatter detection system based on a high-resistivity multielement two-dimensional (2-D) Si-pad array, a multichannel low-noise pulse processing front-end electronics chip, the XA1.3, and a new DAQ system. X-rays in the energy range of 17-45 keV can be detected with a FWHM energy resolution of 1-3 keV. Results on the characterization and optimization of the detector-readout electronics-DAQ system and its performance to measure diffraction signatures of most commonly used breast-equivalent materials of interest are presented.
|Item Type:||Conference or Workshop Item (UNSPECIFIED)|
|Additional Information:||Darambara, D. G., Sellin, P. J., and Maehlum, G. (2004). A Novel Detection System Consisting of a Large-Area Sensor and a Multicell Si-Pad Array Operated in Spectroscopic Mode for X-Ray Breast Imaging. IEEE Transactions on Nuclear Science, Vol. 51, No. 4, pp. 1641-1647. ©2004 IEEE. Personal use of this material is permitted. However, permission to reprint/republish this material for advertising or promotional purposes or for creating new collective works for resale or redistribution to servers or lists, or to reuse any copyrighted component of this work in other works must be obtained from the IEEE.|
|Divisions:||Faculty of Engineering and Physical Sciences > Physics|
|Deposited By:||Mr Adam Field|
|Deposited On:||27 May 2010 15:37|
|Last Modified:||24 Oct 2012 10:17|
Repository Staff Only: item control page